Spore test with results in 3 hours!
Pack x 10 equal units.
Warning: Last items in stock!
3H biological indicator for steam sterilization
Bacteriological medium whose survival and death is calculated according to USP, AAMI, ISO 11138 and all appropriate sub-guidelines.
For the correct control of sterilization, it is recommended to use a biological control in each load, provided that implantable products are made, in the rest of the cases according to the protocols of the center.
Pack x10 equal units.
Activate by pressure (manually or with tweezers) the center of the indicator body until the vial containing the culture broth breaks. Put this in contact with the spore strip that is in the vial base and place in the incubator. Incubate immediately after activation.
The control (when still raw) will provide a positive fluorescence result (red or + light). The correctly processed 3H Biological Indicator will read negative fluorescence (green or - light) after 3 hours of incubation. A negative reading indicates an acceptable sterilization process. Processed 3H Biological Test results are not valid until the positive control reads positive fluorescence (red or + light).